DI 040050 : Nanotorque V2.0 Rotary chair
DI 040050 : Nanotorque V2.0 Rotary chair, programmed, with reclining backrest to the horizontal
Terms and Conditions
30-day money-back guarantee
Shipping: 2-3 Business Days
The rotary chair is used for vestibular examination in the clinic and the practice.
State-of-the-art microprocessor technology enables easy operation and ensures high accuracy of freely programmable parameters.
All movements are freely programmable. The rotating chair is fully compatible with the software Disoft II by Difra. The software, in addition to the controller, also records and analyzes the nystagmus data. The movement of the chair is also recorded.
With a comfortable and fully recline seat, the NanoTorque can be used to perform also other test like caloric or positional and Dix-Hallpike tests.
The chair is equipped with slip rings for connection to Instrumentation Difra VNG devices and on request for Instrumentation Difra ENG device.
The following examination tests enables to monitor the VOR function of the patient,
Rotational Sinus test
Velocity Step Test
Automatic positioning of the chair before testing (individually programmable for each test run).
Other possible versions:
Minitorque: standard version (with direct drive motor).
Megatorque: with electrically adjustable off verticalaxis rotation. Adjustment range of the tilt angle from 0 ° to 15 °.
Gigatorque: with electrically adjustable excentric position for unilateral otolith irritation
Mains voltage: 240 Volt 50/60 Hz
Power consumption: approx. 500 watts
Drive: motor with 3-fold V-belt
Max. speed: 200°/sec. (if the chair is anchored in the ground) - 100°/ sec (without anchoring)
Max. acceleration: 100°/ sec²
Max. load (patient): 150 Kg
Slip rings: 21 (all are used)
Footprint: approx. 1300 x 1300 mm
Safety zone: 2000 x 2000 mm
Weight: approx. 125 Kg
Digital output: for recording the chair movement
Connectable devices: NysStar I (VNG 2 channel), NysStar II (VNG 4 channel), HeadStar (vHit / VNG) and Ideas III on request (ENG)
Nanotorque meets EN 60601-1 Standard and Directive 93/42/EEC.
Once the user has seen at least one product this snippet will be visible.